Smart Sense Infrared Thermometer Non Contact
The most basic design consists of a lens to focus the infrared (IR) energy on to a detector, which converts the energy to an electrical signal that can be displayed in units of temperature after being compensated for ambient temperature variation. Infrared Thermometers allow users to measure temperature in applications where conventional sensors cannot be employed. Specifically, in cases dealing with moving objects ( i.e., rollers, moving machinery, or a conveyor belt), or where non-contact measurements are required because of contamination or hazardous reasons (such as high voltage), where distances are too great, or where the temperatures to be measured are too high for thermocouples or other contact sensors.
|Accuracy(In the condition of 23°C±3°C)|| 100℃(212℉) to 600℃(1112℉)±2℃or±2%|
0℃(32℉) to 100℃(212℉) ±2℃or±2%
-50℃(-58℉)to 0℃(30℉) ±3℃or ±3%
whichever is greater
|Repeatability||1% of reading or 0.1°C|
|Response time||500 mSec, 95% response|
|Spectral response||(8-14)um & 500ms|
|Emissivity||0.10 to 1.00 adjustable (pre-set 0.95)|
|Ambient operating range||0 to 40°C (32°Fto 104°F)|
|Relative humidity||10-95% RH noncondensing|
|Storage temperature||-20 to 60°C (-4°Fto 140°F)|
|Weight||170g without battery|
|Distance to Spot Size||12:1|
|Data Hold Function||V|
|Laser Target Pointer Selection||V|
|Backlight Display Selection||V|
|Auto Power Off||V|
|Power||9v Alkaline or NiCd battery (not inlcude)|
Emissivity : Most organic materials and pa-inted or oxidized surfaces have an emissiv-ity of 0.95 (preset in the unit). Inaccurate readings will result from measuring shiny or polished metal surfaces. To compensate, cover the surface to be measured with mas-king tape or flat black paint. Measure the tape or painted surface when the tape or painted reach the same temperature as the material underneath.
Diagram and LCD display:
Compact, rugged and easy to use. Just aim and push the button,
read current surface temperature in less than a second. Safely
measure surface temperature of hot, hazardous or hard-to-reach
objects without contact.
(2) Laser / back light button: when back light turn
(3)-(6) key functions:
difference of current reading.
When reading over trigger, LCD display HI icon with BiBi audio sounds.
| LCD display:|
A data hold icon
B scanning icon
C laser on icon
D back light on icon
E battery power icon
F measuring unit
G measuring reading
Distance & spot size: As the distance from the object increase,
Make sure the target is larger than the unit's spot size. The detected parts diameter need up to 1.5cm. But the best testing distance is 20cm.when detected parts size is 10*10cm
hole is the main Parts measure the temperature
2. Do not point laser directly at eye or in-directly off reflective surface.