Digital Infrared Thermometer LDC IR Display -32~400C Non Contact Gun-Type infrared thermometer high accurary
Digital Infrared Thermometer LDC IR Display -32~400C Non Contact Gun-Type infrared thermometer high accurary are safely and quickly used for measuring the temperature of the object's surface without contact. The most basic design consists of a lens to focus the infrared (IR) energy on to a detector, which converts the energy to an electrical signal that can be displayed in units of temperature after being compensated for ambient temperature variation.
Digital Infrared Thermometer LDC IR Display -32~400C Non Contact Gun-Type infrared thermometer high accurary, rugged and easy to use.
ust aim and push the button, read current surface temperature in less than a second.
Safely measure surface temperature of hot, hazardous or hard-to-reach objects without contact.
|Temperature range||-32 °C to 400 °C (-26°F to 752 °F )|
|Accuracy(In the condition of 23°C±3°C)|| -32℃(-26℉) to 0℃(32℉) ±3℃|
0℃(32℉)to 100℃(212℉) ±2℃
100℃ hereinbefore ±2℃or±2%
whichever is greater
|Repeatability||1% of reading or 0.1°C|
|Response time||500 mSec, 95% response|
|Ambient operating range||0 to 40°C (32°Fto 104°F)|
|Relative humidity||10-95% RH noncondensing|
|Storage temperature||-20 to 60°C (-4°Fto 140°F)|
|Weight||130g without battery|
|Distance to Spot Size||12:1|
|Data Hold Function||V|
|Laser Target Pointer Selection||V|
|Backlight Display Selection||V|
|Auto Power Off||V|
|Power||2 x AAA battery or NiCd battery (not inlcude)|
Diagram and LCD display:
| 1. Trigger: When pulling the trigger, |
LCD display reading W/ SCAN.
Release the trigger, display reading
W/ HOLD icon for 7s (approx).
Built-in 20s auto power off.
2. Laser on/off button
3. Celsius / Fahrenheit switch button
4. Back light on/off: When back light on,
any operations will keep back light for 10s
6. Battery door: When replacing batteries,
please pull the battery door.
| A data hold icon|
B scanning icon
C laser on icon
D back light on icon
E battery power icon
F measuring unit
G measuring reading
Red laser point only position the general
direction, the detection hole is the main parts measure the
Quick Start Instruction:
|Step 1||Press battery door clip, install battery correctly. Pull the trigger, LCD display reading& battery icon. Release the trigger and the reading will hold for 7 seconds.|
|Step 2||Locating a hot spot: To find a hot spot aim the thermometer outside the area of interest, then scan across with up and down motions until you locate the hot spot.(please turn on the laser for accurate measuring)|
|Instructions||Compact, rugged and easy to use. Just aim and push the button, read current surface temperature in less than a second. Safely measure surface temperature of hot,hazardous or hard-to-reach objects without contact.|
|How it works||Temperature of an object, The optical detector sense the object s emitted energy with different wavelength. It is collected and focus onto a detector. The electronics system translated the information into a temperature reading which is displayed on the unit.|
|Cautions|| Infrared thermometer should be protected from the following:|
--EMF (electro-magnetic fields) from welding equipment, inductive heaters.
--Thermal shock (cause by large or abrupt ambient temperature changes allow 30 minutes for unit to stabilize before use).
-- do not leave the unit on or near objects of high temperature.