Digital non-contact IR gun infrared thermometer 18℃~1650℃ 0℉~3002℉ infrared temperature meter Thermometer
Digital non-contact IR gun infrared thermometer 18℃~1650℃ 0℉~3002℉ infrared temperature meter Thermometer, rugged and easy to use. Just aim and push the button, read current surface temperatures in less than a second. Safely measure surface temperatures of hot, hazardous or hard-to-reach objects without contact.
|Temperature range||18℃~1650℃ 0℉~3002℉|
|Accuracy(In the condition of 23°C±3°C)|| -50℃(-58℉) to -31℃(-23℉) ±5℃|
-30℃(-23℉) to -1℃(31℉) ±3℃
0℃(32℉)to 100℃(212℉) ±2℃
100℃ hereinbefore ±2℃or±2%
whichever is greater:25℃±3℃
|Repeatability||1% of reading or 0.1°C|
|Response time||500 mSec, 95% response|
|Spectral response||(8-14)um & 500ms|
|Emissivity||0.10 to 1.00 adjustable (pre-set 0.95)|
|Ambient operating range||0 to 40°C (32°Fto 104°F)|
|Relative humidity||10-95% RH noncondensing|
|Storage temperature||-20 to 60℃ (-4~140℉)≤85%RH, not including battery|
|Low battery indication||V|
|Weight||480g without battery|
|Dimensions||220 x 134 x 60mm|
|Distance to Spot Size||50:1|
|Read out the stored data||V|
|Clearing the stored data||V|
|Laser Target Pointer Selection||V|
|Backlight Display Selection||V|
|Auto Power Off||V|
|Power||9v Alkaline or NiCd battery (not inlcude)|
Diagram and LCD display:
1. LCD display
| LCD display:|
A: Measuring reading
B: Temperature unit
C: Laser point ON icon
D: Backlight ON icon
E: Battery power indication icon
F: Scanning icon G: Data hold icon
H: Mode / emissivity indicator
Distance & spot size: As the distance from the object increase, the spot size of measuring area becomes larger.
Field of view:
Make sure the target is larger than the unit's spot size. The detected parts diameter need up to 1.5cm. But the best testing distance is 20cm.when detected parts size is 10*10cm
Quick Start Instruction:
|Step 1||Press battery door clip, install battery correctly. Pull the trigger, LCD display reading& battery icon. Release the trigger and the reading will hold for 7 seconds.|
|Step 2||Locating a hot spot: To find a hot spot aim the thermometer outside the area of interest, then scan across with up and down motions until you locate the hot spot.(please turn on the laser for accurate measuring)|
|How it works||Infrared thermometer measure the surface temperature of an objec,t. The units optical system sense the object s emitted energy with different wavelength. It is, collected and focus onto a detector. The unit s electronics system translated the information into a temperature reading which is displayed on the unit.|
|Cautions|| Infrared thermometer should be protected from the following:|
--EMF (electro-magnetic fields) from welders, induction heaters.
--Thermal shock (cause by large or abrupt ambient temperature changes allow 30 minutes for unit to stabilize before use).
--Do not leave the unit on or near objects of high temperature.
Do not point laser directly at eye or in-directly off reflective surface.